factory low price Covid Saliva Home Care Antigen - Testsealabs COVID-19 ANTIGEN RAPID TEST KIT（Swab） – Testsealabs
factory low price Covid Saliva Home Care Antigen - Testsealabs COVID-19 ANTIGEN RAPID TEST KIT（Swab） – Testsealabs Detail:
1.ISO13485&ISO9001, CE, TGA, EU HSC, MHRA, BfrAm, PEI Approved from Original Manufacturer
2. Nasopharyngeal Swab/Oropharyngeal Swab/Nasal Swab Sample
3. Instant Result in 10 minutes, faster than PCR test
4. Easy to use, no equipment required , no PCR Lab required
5: Extraction Tube with buffer, more convinently
6: Room temperature storage(4°C-30°C)
Covid-19 Antigen Rapid Test kit-Nasopharyngeal/Oropharyngeal Swab
Covid-19 Antigen Rapid Anterior Nasal Swab Test kits
|Product Name||Sample Type||Storage Temperature||Packing Specification|
|COVID-19 ANTIGEN TEST KIT(Swab)||Nasopharyngeal Swab/Oropharyngeal Swab/Nasal Swab||4°C-30°C||1T/BOX
Take nasopharyngeal swab for example.
①Peel off aluminum foil seal from the top of the extraction tube containing the extract- ion tube containing the extraction buffer
② Tilt patient’s head back about 70′ . Insert sterilized swab through the nostril parallel to the palate.
③Gently rub and roll the swab, and leave swab in place for several seconds to absorb secretions.
④Slowly remove swab while rotating it.
⑤Insert the swab specimen into the extraction tube. Roll the swab at least 5 times and leave the swab in the extraction tube for one minute.
⑥Remove the swab while squeezing the sides of the tube to extract the liquid from the swab.
⑦Cover the extraction bube with a dropper tip tightly.
⑧ Transfer 3 drops（approximately 100uL）to the specimen well of the test cassette.
⑨Interpret the test results at 15 minutes. Do not read results after 20 minutes.
● Extraction buffer
● Extraction Tube
● Package insert
● Test devices
Materials required but not provided Timer
Allow the test, specimen, buffer and/or controls to reach room temperature 15-30℃ (59-86℉) prior to testing
Positive: Two lines appear. One line should always appear in the control line region(C), and another one apparent colored line should appear in the test line region.
NOTE: The intensity of the color in the test line region (T) will vary based on the amount of COVID- 19 antigen present in the sample. So any shade of color in the test region (T) should be considered positive
Negative: One colored line appear in the control region(C).No apparent colored line appear in the test line region.
Invalid: Control line fail to appear. Insufficient specimen volume or incorrect procedural techniques are the most likely reasons for control line failure. Review the procedure and repeat the test with a new test device. If the problem persists, discontinue using the test kit immediately and contact with your local distributor.
Product detail pictures:
Related Product Guide:
Attaining consumer satisfaction is our company's purpose without end. We will make wonderful endeavours to produce new and top-quality merchandise, satisfy your exclusive requirements and supply you with pre-sale, on-sale and after-sale services for factory low price Covid Saliva Home Care Antigen - Testsealabs COVID-19 ANTIGEN RAPID TEST KIT（Swab） – Testsealabs, The product will supply to all over the world, such as: Stuttgart, Austria, Amsterdam, We'd like to invite customers from abroad to discuss business with us. We can present our clients with high quality products and excellent service. We're sure that we'll have good cooperative relationships and make a brilliant future for both parties.
By May from Canada - 2018.12.28 15:18
We are a small company that has just started, but we get the company leader's attention and gave us a lot of help. Hope we can make progress together!
By Freda from Mombasa - 2018.11.06 10:04